I.Introduction

KC901V is a universal radio frequency multimeter, of which main function is single-port vector network analyzer (VNA, S11) and supports vector transmission test (S21 scalar and vector) simultaneously, simple spectrometer, field intensity meter, radio frequency signal source, audio signal source and other functions. KC901V is professional and practical, and even includes batteries that support 4 hours of continuous operation, its volume is only 1cubic decimeter and its weight is only 1kg, bringing a brand-new free experience to users.

II.Main Features *

9KHz-6.8 GHz Frequency Range

1Hz Frequency stepping

All-digital Intermediate Frequency

Reliable accuracy and stability

Strong anti-interference ability

Rich function and portable design



III.Main Function

Transmission Test (amplitude: debug filter, amplifier, verify antenna directivity, phase: provide basic phase change trend)



Reflection Test (vector: debug impedance matching, check antenna feeder system)

Spectrum display and field strength observation ** (check the transmission performance of the radio station and find out interference sources)

Audio Signal Source

Radio Frequency Signal Source (Carrier)

IV.Recommended Application

In particular, it is applicable to:

Standing wave, impedance, return loss and fault location of antenna;

Test the frequency spectrum and field strength of carrier signals, including searching for interference sources

Preliminary debugging of FM / AM receiver sensitivity

Test the isolation of the shielded room

V.Scope of application:

Television Transmitting Station

Indoor distribution engineering

Wireless Internet of Things Project

Professional Communications

Confidentiality, Security

Education

Unmanned Aerial Vehicles, Remote Telemetry and Refusal to Stop

Satellite Earth Station, Radio Astronomy

Installation and maintenance of nuclear magnetic resonance equipment

Secondary Development (Interface Protocol Disclosure)

(The applicable fields are for reference only. Please consult customer service according to the different scenarios.)

VI.Not applicable:

Spectrum and field strength functions are not suitable for measuring pulse signals (such as radar) and frequency hopping signals. In some cases, it can be observed qualitatively, and experienced engineers can still complete the test, but we do not provide any guarantee.

No test should have strong interference signals in the range of 109-110MHz. Some broadcast transmitters have 109 MHz spurs. Measuring the antenna near the transmitter will affect the accuracy of the instrument. This type of test is suitable for KC901S.

For special purposes, please contact customer service for confirmation. Since 7-day return is supported, you can also purchase and try it when necessary.

*Subject to published technical parameters, **Informal functions, KC901 is a network analyzer, and other functions are practical tools derived from the network analyzer architecture for reference only.

Contact us:

Shaanxi Vision Radio and Television Equipment Co.,Ltd

Wang

WhatsApp: 0086 13087559846