3.5MHz R60 convex probe (2.0/2.5/3.5/5.0MHz) - $ 4,999.00 USD 7.5MHz L40 linear probe (6.0/6.5/7.5/9.0MHz) - $ 4,999.00 USD 6.5MHz R20 micro-convex probe(5.0 / 6.5 /7.5 / 9.0 MHz) - $ 4,999.00 USD 6.5MHz linear rectal probe (4.5 / 5.5 / 6.5 / 7.5MHz) - $ 4,999.00 USD With Two Probes/Specify Which at Checkout - $ 5,999.00 USD

Probe Options 3.5MHz R60 convex probe (2.0/2.5/3.5/5.0MHz) 7.5MHz L40 linear probe (6.0/6.5/7.5/9.0MHz) 6.5MHz R20 micro-convex probe(5.0 / 6.5 /7.5 / 9.0 MHz) 6.5MHz linear rectal probe (4.5 / 5.5 / 6.5 / 7.5MHz) With Two Probes/Specify Which at Checkout

Affordable Handheld Color Doppler Ultrasound

DCU-12 Vet is the most affordable full digital color doppler system that you can buy. It is easy to use and has excellent image quality. The DCU-12 Vet Ultrasound provides reliable performance, includes an abundance of measurement software. Provides advanced practical clinical solutions for ultrasound diagnosis of cattle, dogs, horses, cats, goats, sheep, and pigs.

Standard Configuration

Main Unit

Choice of 4 probes:

3.5MHz R60 convex probe (2.0/2.5/3.5/5.0MHz)



7.5MHz L40 linear probe (6.0/6.5/7.5/9.0MHz)



6.5MHz R20 micro-convex probe(5.0/6.5/7.5/9.0 MHz)



6.5MHz linear rectal probe (4.5/5.5/6.5/7.5MHz)

Features

Specific Veterinary Software Packages

Transmitted Signal Accuracy Control

i-Image, Image Optimization Imaging Enhancement

Adaptive Color Artifact Removal Technique

Linear Probe Independent Angle Deflection

i-Station, Integral Working Station

Phase Enhance Harmonic

Accurate Vessel Imaging

Convenient and Practical Data Manager

Abundant Measuring Software

General Description

Built-in embedded A8 platform, totally against virus invasion, with high reliability

Operating system: Linux

Digital multi-beam forming technology, precision delay-by-point dynamic receive focus, ultra-wideband imaging, adaptive image optimization, adaptive angiography, adaptive Doppler imaging

Ultra-wideband fully digital beam former: Dynamic Focus

M-mode imaging

Visual adjustable dynamic range

Gray-scale imaging: ≥256 levels, with excellent contrast resolution and the audience subtle and uniformity

Technical Parameters

Gray Scale: 256 levels

Color Scale: 256 levels

Display: 10.4’ Flicker-inch high-resolution LCD dedicated medical color display

Power Supply: 100-240V～ 1.2-0.6A

Fre: 50-60Hz

Adapter Output: DC12.8V 3.0A

Power Consumption: ≤100VA

Main Unit Dimension: approximately 256mm × 150mm × 326mm (L×W×H)

Main Unit Net Weight: 10 lbs. (without accessories)

Multimedia & Peripheral Equipment

V ideo printer

i-Station integrated workstations, to achieve store images, report generation and cloud printing capabilities

U disk (file management support, software upgrades, and one-key store function), DICOM interface, easy to doctor's data management, remote transmission

Dual TV output: PAL/NTSC

Operating Modes

B, B/B, 4B mode

M, B/M mode

CFM

PDI

PW

THI

Operating System Setup

Basic Setting: Interface language, time, date and date format, screen-savers time settings, key tone setting, standard setting, font brightness setting.

Formula Preset: gestational formula for cattle, dogs, horses, cats, goats, camels, sheep and pigs.

Notes Preset: disease, routine, abdominal, obstetric, cardiac, urology, superficial seven kinds, can be preset for each common noun, and can be added, modified and edited according to the actual requirements, very user-friendly

Stored Presets: Photo storage format selection: BMP, PNG format (image compression format).

Password Preset

Main Functions

System Preset

I-User-defined, to store the setting while he used the main unit to save time to set again

Probe switch

Mode switch

Magnification / Depth selection

Frequency switch

Frame switch

ID

Compression curve

IP

Focus switch

Up/down, left/right reverse

TGC, Gain & Dyn adjustment

i-Image optimization imaging technology

Color frequency, PRF, wall filter adjustment

Deflection angle, sample volume, linear color deflection, velocity scale, baseline function can be adjusted

Adjustable harmonic

B mode normal measurement, M mode routine measurements, CFM mode routine measurements, PDI mode routine measurements, PW mode measurements; and abdominal measurement applications, obstetrics application measurement, measuring cardiac applications, measuring urinary system applications, superficial organ application measurement

Sound power adjustable

Puncture guiding function

Gravel location positioning function

Built-in disk storage 16G, mass storage images, reports, videos, permanent storage, (BMP format picture ≥ 10 thousands, PNG format images ≥100 thousands)

Image management function

Mass video storage and replay (Under B mode: 380, under M mode: 4000 ）

Body marks

Note input

Patient info input

Report generation

One key store

One key to get report

One key to transfer image to Network Printer to print the image

LCD brightness adjustable

Energy saving

Image Pretreatment

Total Gain: 0～127 dB

TGC: 8 stages

Power Output: 25%, 50%, 75%, 100%

Digital Channels: 32

Image Treatment

IE: 0～3

Frame Correlation: 0～3

IP: 0～7

Compression Curve: 0～7

B Mode Scanning Parameters

B Sound Output: 25%, 50%, 75%, 100%

B Gain: 0～127

Depth: 32～252 mm (35 scales)

Dyn: 30～100dB

Frequency: 4 scales

Harmonic: 2 scales

M Mode Scanning Parameters

M Sound Output: 25%, 50%, 75%, 100%

M Gain: 0～127dB

Depth: 32～252 mm (35 scales)

M Dyn: 27～100dB

M Scanning Speed: 1～8 adjustable

M Sampling Line Visual & adjustable

M Scanning Mode: scrolling & scanning

Frequency: 4 scales

Harmonic: 2 scales

CFM Mode, PDI Mode

Output: 25%, 50%, 75%, 100%

CFM Gain: 0～127dB

Color Frequency: 2 scales

Max PRF: 5kHz

Min PRF: 35kHz

CFM Wall Filter: 8 step 50-350Hz

Sensitivity: 6 kinds

Color Afterglow: 4 kinds

Color Threshold: 6 kinds

Color Balance: 16 steps

Color Smoothing: 3 steps

Artifact Removal: 2 steps

Linear Color Deflection: Adjustable

Image Optimization: 12 scales

PW Mode

Sound Output: 25%, 50%, 75%, 100%

PW Gain: 0 ～ 127dB

127dB PW Dyn: 13 scales 0-120

PW Frequency: 2 scales

PW Wall Filter: 8 steps 50-350Hz

Noise Reduction: 4 scales

Sample Volume Width: 5-20mm

Angle Correction

Basic Line Adjustment: 7 steps

Max PRF (speed): 5kHz

Min PRF (speed): 35kHz

Speed Ruler: 8 steps

Volume: adjustable

Linear Color Deflection: adjustable

Measurement & Calculation

Routine Measurement: measurement software package & report for B mode, Ｍ mode, D mode

mode, D mode OB software for Cattle, dogs, horses, cats, goats, camels, sheep, and pigs

Software package for urological measurement & analysis

Software package for superficial organs measurement & analysis

Software package for cardio measurement & analysis

Reports

OB/GYN

Cardiology

Urological

Marks

127 kinds of body marks with arrow position

Note mark

Text input

Arrow mark

Optional parts