Vet Multi-Parameter Monitor KM-7Vet

Excellent quality multi-parameter veterinary patient monitor KM-7Vet. Features specific vet software for small and large animals. This equipment can monitor such parameters as ECG, RESP, SpO2, NIBP, and Dual- channel TEMP. It integrates parameter measuring module, display and recorder in one device to form a compact and portable device. At the same time, its built-in replaceable battery provides convenience and is suitable for use on-the-go.

Features

7-inch high-resolution color screen

Exquisite design suitable to move operation

Various interface: standard, trend screen, oxycrg, NIBP list, big font screen

Real-time S-T segment analysis, pace-maker detection

Indication of incorrect manipulation and malfunction analysis

Patient info input management function

Capture dynamic waveforms

Built-in rechargeable lithium battery, 5 hours working capacity

Standard Configuration

SPO2

PR

NIBP

ECG

RESP

TEMP

Optional Configuration

ETCO2

Parameters

NIBP: Method: Oscillometry Modes: Manual/Auto/Continuous; Auto measure time: Adjustable Measurement range:10-270mmHg

ECG: Lead mode: 3-lead or 5-lead Lead selection: I; II; III, avR; avF; V Heart rate range: Adult:15-300bpm; Pediatric/Neonatal:15-300bpm ECG waveform: 2 channels; Accuracy: ±1 bpm or ±1%,which is greater S-T segment detection;measurement range: -2.0mv~+2.0mv; Alarm:Yes,audible and visual alarm, alarm events review

SPO2: Measurement range:0-100% Accuracy:70-100%,±2%,0-69%,unspecified; Auto measure time: adjustable

Pulse Rate: Range: 20-254bpm; Accuracy: ±3bpm

Respiration: Method: RA-LL impedance; Range: Adult:0 120rpm; pediatric/neonatal: 0-150rpm; Apnca alarm: Yes

Temperature: Range: 0-500; Accuracy: ± 0.10; Channel: Dual-channel

Main Unit: ECG cable, Spo2 probe, TEMP Probe, NIBP Tube Adult cuff, Power cable, Battery, User manual

Accessories: