Home » Our Products » General Equipments » General Equipments

General Equipments









LYOPHILIZER (FREEZE DRYER)

Housed in a super structure steel frame it is mounted on swivel casters. Complete with Perspex see though lid, gasket and control instrument i.e. Microprocessor Based Digital Temperature controller for measurement of Temp. and Electronic Vacuum Indicator. The unit made out of hermetically sealed compressor and with 100 liters/ min direct motor drive rotary vane type vacuum pump of double stage With Air ballast having ultimate vacuum of less than 50 microns in the system and 10 microns on pump Head. Condenser Trap: All non-magnetic Stainless steel fabricated, argon welded, dished-bottom for complete effluent removal. Refrigeration coils made of copper bonded to outside of condenser heavily Insulated, up to (-) 40°C, at ambient temperature of 20°C, is maintained through a MICROPROCESSOR BASED DIGITAL TEMPERATURE INDICATOR. Vacuum Drum has 6 or 12 port manifold (main driving Chamber) made of stainless steel non-magnetic grade with Perspex top cover. (CFC FREE AND PUFF INSULATED)

TECHNICAL SPECIFICATIONS & ORDERING INFORMATIONS





Capacity 3.25 Ltrs 5.25 Ltrs Vacuum Drying Drum Size 22.5cm (dia) x 25.5cm (ht) 29 Ltr. MOC (Body) CRC duly power coated on caster wheels MOC (Main Drying Chamber) Stainless Steel (Non-Magnetic) Temperature Range -40oC (at ambient temperature of 20oC) Controller Microprocessor based temperature control system with inbuilt HI/LOW temperature deviation alarm Vacuum Probe Electric vacuum transducer Vacuum Indicator Digital vacuum indicator Vacuum Seals Made of Silicon rubber Ultra Ultimate Vacuum (Pump Head) 1x 10-3 torr No. of Draying Ports 6 12 Vacuum Pump 100 liters/ min direct motor drive rotary vane type vacuum pump of double stage with Air ballast having ultimate vacuum of 1x10-3 torr on pump head Heat Rack (Option) With 2/3 individually heated racks for accelerated freeze drying of small samples in bulk (Tray, Petri Dishes, Beakers etc)







