Manual EN

Translations Schematics Code Hex BOM Interface Videos Pictures FAQ Forums IN DEVELOPMENT

The Oscilloscope Watch has all the features of a modern watch (time, calendar, alarm, etc...) combined with all the features of the popular Xprotolab (Oscilloscope, Waveform Generator, Logic Analyzer, Protocol Sniffer, Frequency Counter). The Xprotolab has been featured in several other online magazines, such as Hack a Day. and Elektor. Features: Mixed Signal Oscilloscope: Simultaneous sampling of analog and digital signals.

Advanced Trigger: Normal / Single / Auto, with rising or falling edge and adjustable trigger level.

Meter Mode: Average, Peak to peak and Frequency readout.

XY Mode (Plot Lissajous patterns or see the phase difference between two waveforms).

Spectrum Analyzer with different windowing options and selectable vertical log.

Horizontal and Vertical Cursors with automatic waveform measurements.

Arbitrary Waveform Generator with Sweep on all parameters.

Display options: Persistence, Different grid options, and more.

Curve tracer function General Specifications: ATXMEGA256A3U 256KB Flash, 16KB SRAM, 4KB EEPROM

Sharp Memory LCD display 1.28", 128x128 pixels

Approximate watch size: 2" x 1.6" x 0.6"

Li-Ion Battery 400mAh

Internal PDI interface.

8 tactile switches

USB connectivity: Windows, Linux, MAC, Android Logic Analyzer specifications: 8 Digital Inputs, 3.3V / 5V

820k ohm input impedance

Maximum sampling rate: 16MSPS

Frequency counter: 16Mhz

Protocol Sniffer: UART, I2C, SPI

Internal pull up or pull down.

Buffer size: 4096 Oscilloscope specifications: 2 Analog Inputs

Maximum Sampling rate: 4MSPS

Analog Bandwidth: 200kHz

Resolution: 8bits

Input Impedance: 1M? 15pF

Buffer size per channel: 4096

Input Voltage Range: -14V to +20V AWG specifications: 1 Analog Output

Maximum conversion rate: 1MSPS

Analog Bandwidth: 50kHz

Resolution: 8bits

Output current > +/- 7mA

Buffer size: 512

Output Voltage: +/- 4V �

